|rainbow® Sensors & Cables||SatShare|
Pronto — with rainbow® technology — for noninvasive and quick spot-checking of total hemoglobin (SpHb®), arterial oxygen saturation (SpO2), pulse rate (PR), and perfusion index (PI)
Your Solution for Total Hemoglobin Spot-checking
The Pronto may provide the following benefits:
- > Facilitates timely patient assessment
- > Easy to use
- > Expands information available to your clinician at time of assessment
Four Simple Steps
Select sensor size
Place sensor on finger
Press SpHb button
Total Hemoglobin Performance
In 492 comparisons of SpHb and invasive hemoglobin (tHb) measurements from a laboratory CO-Oximeter, SpHb accuracy was as follows*:
- • 0.08 g/dL bias
- • 0.96 g/dL ARMS accuracy
- • Below 12 g/dL, 99% of SpHb readings were <2 g/dL of the laboratory tHb value
- • At or above 12 g/dL, 94% of SpHb readings were <2 g/dL of the laboratory tHb value
*Masimo FDA Submission Data
Please note: These results were obtained following a specific protocol, in which all Directions for Use were followed and SpHb measurements were compared to validated laboratory reference measurements. Differences in results may be caused by many factors, such as those explained in Masimo’s Directions for Use. Some independent researchers have conducted their own testing and obtained similar results, while other researchers have reported larger differences when comparing SpHb measurements to laboratory measurements.
SpHb accuracy has been validated on healthy adult male and female volunteers and on surgical patients with light to dark skin pigmentations in the range of 8 g/dL to 17 g/dL. SpHb has not been validated under motion or low perfusion conditions. The SpHb accuracy specification is calculated at one standard deviation, which encompasses approximately 68% of the measurement comparisons to a laboratory reference device. Please contact Masimo for additional testing specifications.
Please note: SpHb accuracy specifications were obtained by following a specific protocol, and SpHb measurements were compared to validated laboratory reference measurements. Differences in results may be caused by many factors, such as those explained in the device instructions for use. Some independent researchers have conducted their own testing and obtained similar results, while other researchers have reported larger differences when comparing SpHb measurements to laboratory measurements.
Caution: Federal (USA) law restricts this device to sale by or on the order of a physician. See instructions for use for full prescribing information, including indications, contraindications, warnings, and precautions.