|rainbow Sensors & Cables||SatShare|
Pronto-7 — with rainbow 4D™ technology — for noninvasive and quick spot checking of total hemoglobin (SpHb®), oxygen saturation (SpO2), pulse rate (PR), and perfusion index (PI)
Your Solution for Hemoglobin Spot-checking
The Pronto-7 may provide the following benefits:
- > Facilitates timely patient assessment
- > Easy-to-use — Improves efficiency
- > Expands information available to your clinician at time of assessment
Four Simple Steps
Select sensor size
Place sensor on finger
Press test button
Automatically saved — option to print,
e-mail, or manually record
SpHb Analytical Performance
In 11,335 comparisons of SpHb and invasive hemoglobin (tHb) measurements from a laboratory reference device, SpHb accuracy was as follows:*
- • -0.02 g/dL bias
- • 0.99 g/dL ARMS accuracy
- • Between 6 - 12 g/dL, greater than 95% of SpHb readings were within 2 g/dL of the laboratory tHb value
- • Between 12 - 18 g/dL, greater than 95% of SpHb readings were within 2 g/dL of the laboratory tHb value
*Masimo FDA 510(k) Submission Data using normal sensitivity mode.
Please note: These results were obtained in normal sensitivity mode following a specific protocol, in which all Directions for Use were followed and SpHb measurements were compared to validated laboratory reference measurements. Differences in results may be caused by many factors, such as those explained in Masimo’s Directions for Use. Some independent researchers have conducted their own testing and obtained similar results, while other researchers have reported larger differences when comparing SpHb measurements to laboratory measurements.
Pronto-7 In-service Training
Prior to using this device, the user should read and understand the Operator’s Manual and Directions for Use. Laboratory diagnostic tests using blood samples should be conducted prior to clinical decision making to completely understand the patient’s condition. Comparisons between SpHb measurements and laboratory diagnostic hemoglobin measurements may be affected by sample type, collection technique, physiological, and other factors.
Caution: Federal law restricts this device to sale by or on the order of a physician